Human IL-2R alpha ELISA Kit (HUES02134)
- Product Type:
- ELISA Kit
- 96 Assays
- ELISA Type:
- Tested Sample Types:
- Serum, plasma and other biological fluids
|Detection Range:||93.75-6000 pg/mL|
|Sample Volume Required Per Well:||100µL|
|Sample Type:||Serum, plasma and other biological fluids|
|Specificity:||This kit recognizes Human IL2R alpha ELISA Kit in samples. No significant cross-reactivity or interference between Human IL2R alpha ELISA Kit and analogues was observed.|
This ELISA kit uses Sandwich-ELISA as the method. The micro ELISA plate provided in this kit has been pre-coated with an antibody specific to Human IL2R alpha. Standards or samples are added to the appropriate micro ELISA plate wells and combined with the specific antibody. Then a biotinylated detection antibody specific for Human IL2R alpha and Avidin-Horseradish Peroxidase (HRP) conjugate are added to each micro plate well successively and incubated. Free components are washed away. The substrate solution is added to each well. Only those wells that contain Human IL2R alpha, biotinylated detection antibody and Avidin-HRP conjugate will appear blue in color. The enzyme-substrate reaction is terminated by adding Stop Solution and the color turns yellow. The optical density (OD) is measured spectrophotometrically at a wavelength of 450 nm ± 2 nm. The OD value is proportional to the concentration of Human IL2R alpha. The concentration of Human IL2R alpha in samples can be calculated by comparing the OD of the samples to the standard curve.
|UniProt Protein Function:||IL2RA: Receptor for interleukin-2. Non-covalent dimer of an alpha and a beta subunit. IL2R exists in 3 different forms: a high affinity dimer, an intermediate affinity monomer (beta subunit), and a low affinity monomer (alpha subunit). The high and intermediate affinity forms also associate with a gamma subunit.|
|UniProt Protein Details:|
Protein type:Membrane protein, integral
Chromosomal Location of Human Ortholog: 10p15. 1
Cellular Component: plasma membrane
Molecular Function:interleukin-2 binding; Ras guanyl-nucleotide exchange factor activity
Biological Process: apoptosis; cell proliferation; cell surface receptor linked signal transduction; immune response; inflammatory response; MAPKKK cascade; regulation of T cell tolerance induction
Disease: Diabetes Mellitus, Insulin-dependent, 10; Interleukin 2 Receptor, Alpha, Deficiency Of
|NCBI Summary:||The interleukin 2 (IL2) receptor alpha (IL2RA) and beta (IL2RB) chains, together with the common gamma chain (IL2RG), constitute the high-affinity IL2 receptor. Homodimeric alpha chains (IL2RA) result in low-affinity receptor, while homodimeric beta (IL2RB) chains produce a medium-affinity receptor. Normally an integral-membrane protein, soluble IL2RA has been isolated and determined to result from extracellular proteolyisis. Alternately-spliced IL2RA mRNAs have been isolated, but the significance of each is presently unknown. Mutations in this gene are associated with interleukin 2 receptor alpha deficiency. [provided by RefSeq, Nov 2009]|
|NCBI GenInfo Identifier:||124317|
|NCBI Gene ID:||3559|
|NCBI Accession:||P01589. 1|
|UniProt Secondary Accession:||P01589,Q5W007,|
|UniProt Related Accession:||P01589|
|Molecular Weight:||30,819 Da|
|NCBI Full Name:||Interleukin-2 receptor subunit alpha|
|NCBI Synonym Full Names:||interleukin 2 receptor subunit alpha|
|NCBI Official Symbol:||IL2RA|
|NCBI Official Synonym Symbols:||p55; CD25; IL2R; IMD41; TCGFR; IDDM10|
|NCBI Protein Information:||interleukin-2 receptor subunit alpha|
|UniProt Protein Name:||Interleukin-2 receptor subunit alpha|
|UniProt Synonym Protein Names:||TAC antigen; p55; CD_antigen: CD25|
|UniProt Gene Name:||IL2RA|
As the OD values of the standard curve may vary according to the conditions of the actual assay performance (e. g. operator, pipetting technique, washing technique or temperature effects), the operator should establish a standard curve for each test. Typical standard curve and data is provided below for reference only.
Intra-assay Precision (Precision within an assay): 3 samples with low, mid range and high level Human IL2R alpha ELISA Kit were tested 20 times on one plate, respectively.
Inter-assay Precision (Precision between assays): 3 samples with low, mid range and high level Human IL2R alpha ELISA Kit were tested on 3 different plates, 20 replicates in each plate.
|Intra-assay Precision||Inter-assay Precision|
|C V (%)||5.67||4.82||5.03||6.45||4.02||3.73|
The recovery of Human IL2R alpha ELISA Kit spiked at three different levels in samples throughout the range of the assay was evaluated in various matrices.
|Sample Type||Range (%)||Average Recovery (%)|
|EDTA plasma (n=5)||84-100||91|
|Cell culture media (n=5)||91-106||97|
Samples were spiked with high concentrations of Human IL2R alpha ELISA Kit and diluted with Reference Standard & Sample Diluent to produce samples with values within the range of the assay.
|Serum (n=5)||EDTA plasma (n=5)||Cell culture media (n=5)|
An unopened kit can be stored at 4°C for 1 month. If the kit is not used within 1 month, store the items separately according to the following conditions once the kit is received.
|Micro ELISA Plate(Dismountable)||8 wells ×12 strips||-20°C, 6 months|
|Reference Standard||2 vials|
|Concentrated Biotinylated Detection Ab (100×)||1 vial, 120 µL|
|Concentrated HRP Conjugate (100×)||1 vial, 120 µL||-20°C(shading light), 6 months|
|Reference Standard & Sample Diluent||1 vial, 20 mL||4°C, 6 months|
|Biotinylated Detection Ab Diluent||1 vial, 14 mL|
|HRP Conjugate Diluent||1 vial, 14 mL|
|Concentrated Wash Buffer (25×)||1 vial, 30 mL|
|Substrate Reagent||1 vial, 10 mL||4°C(shading light)|
|Stop Solution||1 vial, 10 mL||4°C|
|Plate Sealer||5 pieces|
|Product Description||1 copy|
|Certificate of Analysis||1 copy|
- Set standard, test sample and control (zero) wells on the pre-coated plate and record theirpositions. It is recommended to measure each standard and sample in duplicate. Note: addall solutions to the bottom of the plate wells while avoiding contact with the well walls. Ensuresolutions do not foam when adding to the wells.
- Aliquot 100 µL of standard solutions into the standard wells.
- Add 100 µL of Sample / Standard dilution buffer into the control (zero) well.
- Add 100 µL of properly diluted sample (serum, plasma, tissue homogenates and otherbiological fluids) into test sample wells.
- Cover the plate with the sealer provided in the kit and incubate for 90 min at 37 °C.
- Aspirate the liquid from each well, do not wash. Immediately add 100 µL of BiotinylatedDetection Ab working solution to each well. Cover the plate with a plate seal and gently mix. Incubate for 1 hour at 37 °C.
- Aspirate or decant the solution from the plate and add 350 µL of wash buffer to each welland incubate for 1-2 minutes at room temperature. Aspirate the solution from each well andclap the plate on absorbent filter paper to dry. Repeat this process 3 times. Note: a microplatewasher can be used in this step and other wash steps.
- Add 100 µL of HRP Conjugate working solution to each well. Cover with a plate seal andincubate for 30 min at 37 °C.
- Aspirate or decant the solution from each well. Repeat the wash process for five times asconducted in step 7.
- Add 90 µL of Substrate Reagent to each well. Cover with a new plate seal and incubate forapproximately 15 min at 37 °C. Protect the plate from light. Note: the reaction time can beshortened or extended according to the actual color change, but not by more than 30min.
- Add 50 µL of Stop Solution to each well. Note: Adding the stop solution should be done inthe same order as the substrate solution.
- Determine the optical density (OD value) of each well immediately with a microplate readerset at 450 nm.